Biomek i-Series Automated Beckman Coulter AMPure XP PCR Purification SystemBiomek Automated Genomic Sample Prep Accelerates Research
The Beckman Coulter AMPure XP kit is a highly efficient PCR purification system that delivers superior quality DNA with no salt carryover. The AMPure XP protocol utilizes Solid Phase Reversible Immobilization (SPRI) magnetic bead-based technology which requires no centrifugation or filtration. This kit is a staple in NGS sample prep and can be easily used in manual and automated 96- and 384- well formats. Here we will demonstrate automated performance on the Biomek i5 Multichannel 96 Genomics Workstation.
The AMPure XP PCR Purification system is automated on all Biomek platforms and provides:
- Standardized workflow for improved results
- Reduction in costly errors
- Reduced hands-on-time and increased throughput
- Quick implementation with ready-to-implement methods
- Knowledgeable support for reagents, automation and methods all from single vendor
Spotlight: Biomek i5 Multichannel 96 Genomics Workstation
System features deliver reliability and efficiency to increase user confidence and walk-away time
Figure 1. Biomek i5 Multichannel 96 Genomics Workstation with optional enclosure on a Biomek Cart. Large Deck capacity to increase walk away time.
Demonstrated Method Interface (DMI):
Three simple modules that provide the user full instructions to better ensure error-free method setup and provides users maximum flexibility for scheduling their day
1. Biomek Method Launcher (BML)
Secure interface for selecting methods without affecting method integrity and manual control
Figure 2. Biomek Method Launcher provided an easy interface to start the method
Figure 3. Manual Control can be run through the launcher interface
2. Method Options Selector (MOS)
Select run-time options and maximize flexibility in daily scheduling and method execution
Figure 4. Beckman Coulter AMPure XP Method Options Selector showing the different features and run options
3. Guided Labware Setup (GLS)
Generated based on options selected in the MOS, and provides the user specific text and graphical setup instructions with reagent calculation
Figure 5. Guided Labware Setup showing reagent volumes and guides the user for correct deck setup
Fast and efficient process with scalable throughput to maximize yield
Figure 6. Beckman Coulter AMPure XP workflow.
|Major Process Description
|AMPure XP Method
|Hands on Time
|**Time does not include thawing of reagents
Table 1. Beckman Coulter AMPure XP estimated run times on i5 Multichannel 96
Experimental Design and Results
Thermo Marker (500bp) was diluted to 60ng/uL and 50uL was used for AMPure XP cleanup. 5 technical replicates were aliquoted in PCR plate in a checkerboard pattern. After the cleanup on i5 Multichannel, the samples were analyzed on Agilent TapeStation 2200 with High Sensitivity D1000 kit.
Figure 7. Samples were analyzed on Agilent TapeStation using HS D1000 kit. The replicates behaved in a consistent manner and the alternate empty wells show no cross contamination. Recovery of the samples was approximately 95% and gel and trace show no impurity.
Figure 8. Electropherogram showing the sample running around expected size of the marker.
The data shows that Beckman Coulter AMPure XP PCR purification system automated on the new Biomek i-Series Workstation continues the tradition of delivering a high yield of quality DNA.